Turbidity and Free/Total Chlorine Benchtop Meter, EPA Compliant
- HI 83414 features four measuring ranges: ratio, non ratio, free chlorine, total chlorine
- Meets USEPA requirements
- Exclusive chlorine CAL CHECK™ calibration validation
- GLP Features
- Two, three, four or five point turbidity calibration
- USB PC connectivity
- Backlit LCD
- On-screen tutorial and help modes
- Log and recall up to 200 measurements
- Contextual help and tutorial mode
HI 83414 is a highly accurate dual parameter instrument that reflects HANNA’s years of experience. The HI 83414 successfully combines turbidity and colorimetric measurements to test the most important parameters of drinking water: turbidity and free/total chlorine. The meter is specially designed for water quality measurements, providing reliable and accurate readings on low turbidity and chlorine values. The HI 83414 meets and exceeds the requirements of USEPA and Standard Methods both for turbidity and colorimetric measurements.
With the powerful CAL CHECK™ function, reliable performance can be validated at any moment by using the exclusive HANNA ready-made, NIST traceable standards. A one-point calibration can be performed using the same CAL CHECK™ standard.
This instrument features an optical system to guarantee accurate results, assure long term stability and minimize stray light and color interferences. They also compensate for variations in intensity of the lamp for less frequent calibration.
The 525 nm interference filter of the colorimeter assures accurate and repeatable results. Repeatability of the measurements are ensured with 25 mm round cuvettes made from special optical glass.
Turbidity measurements can be made in the 0.00 to 4000 NTU (Nephelometric Turbidity Units) range when ratiometric measurements are used and in the 0.00 to 40.0 NTU range when non ratio method is used. The HI?83414 has an EPA compliance reading mode which rounds the reading to meet EPA reporting requirements. Alternative EBC and Nephelos measuring units are available. Depending on the measured sample and needed accuracy, normal measurement, continuous measurement or signal averaging measurement can be selected.
A two, three, four or five-point calibration can be performed by using the supplied (<0.1, 15, 100, 750 and 2000 NTU) standards. If user prepared standards are used, the calibration points can be modified. Free or total chlorine measurements can be made in the 0.00 to
5.00 mg/L (ppm) range.
HI 83414 features complete GLP (Good Laboratory Practice) functions that allow traceability of the calibration conditions. The last calibration points, time and date can be checked.
This meter also incorporates a user-friendly interface with an easy to understand, graphic LCD. All messages are in plain text making them easy to read. Comprehensive contextual help is available at the press of a button. All messages and help screens are available in several languages. Confirmation and error acoustic signals help the user during instrument operation. Furthermore, a tutorial mode of operation guides the user step by step through the analysis process.
The logging function offers complete measurement information. Up to 200 measurements can be stored in the internal memory and consulted at any time. For further storage or analysis options, data can be downloaded to a PC using the USB port.
HI 83414-01 (115V) and HI 83414-02 (230V) are supplied with sample cuvettes and caps (5), calibration cuvettes for turbidimeter and colorimeter (HI 83414-11), silicone oil (HI 93703-58), cloth for wiping cuvettes, scissors, power cord and instruction manual.
|Range||Non Ratio Mode||0.00 to 9.99; 10.0 to 40.0 NTU;|
0.0 to 99.9; 100 to 268 Nephelos
0.00 to 9.80 EBC
|Ratio Mode||0.00 to 9.99; 10.0 to 99.9; 100 to 4000 NTU|
0.0 to 99.9; 100 to 26800 Nephelos
0.00 to 9.99; 10.0 to 99.9; 100 to 980 EBC
|0.00 to 5.00 mg/L (ppm)|
|Resolution||Non Ratio Mode||0.01; 0.1 NTU; 0.1; 1 Nephelos; 0.01 EBC|
|Ratio Mode||0.01; 0.1; 1 NTU; 0.1; 1 Nephelos; 0.01; 0.1, 1 EBC|
|0.01 mg/L (ppm) from 0.00 to 3.50 mg/L (ppm)|
0.10 above 3.50 mg/L (ppm)
|Turbidity||±2% of reading plus 0.02 NTU (0.15 Nephelos; 0.01 EBC)|
±5% of reading above 1000 NTU (6700 Nephelos; 245 EBC)
|±0.03 mg/L or ±3% of reading|
(whichever is greater)
|Method||Turbidity||Nephelometric method (90°) or Ratio Nephelometric Method (90° & 180°), Adaptation of the USEPA Method 108.1 and Standard Method 2130 B|
|adaptation of the USEPA Method 330.5|
and Standard Method 4500-Cl G.
|Light Detector||Turbidity||silicon photocell|
|silicon photocell with 525 nm narrow band interference filters|
|Standards||Turbidity||<0.1, 15, 100, 750 and 2000 NTU|
|1.00 mg/L free/total chlorine|
|Calibration||Turbidity||two, three, four or five-point calibration|
|Turbidity Range Selection||automatic|
|Repeatability||±1% of reading or 0.02 NTU (0.15 Nephelos; 0.01 EBC) whichever is greater|
|Stray Light||< 0.02 NTU (0.15 Nephelos; 0.01 EBC)|
|Measuring Mode||normal, average, continuous|
|Light Source/Life||tungsten filament lamp / greater than 100,000 readings|
|Display||40 x 70 mm graphic LCD (64 x 28 pixels) with backlight|
|Log Memory||200 records|
|Environment||0°C (32°F) to 50°C (122°F); max 95% RH non-condensing|
|Power Supply||230 V/50 Hz or 115 V/60 Hz 20 W; auto-off after 15 minutes of non-use|
|Dimensions||230 x 200 x 145 mm (9 x 7.9 x 5.7”)|
|Weight||2.5 Kg (88 oz.)|